ASTM D – Standard Test Method for Dielectric Breakdown Voltage of Insulating Oils of Petroleum Origin. Using VDE Electrodes. ▫ IEC Insulating. Page 1. Page 2. Page 3. Page 4. Find the most up-to-date version of ASTM D at Engineering
|Published (Last):||3 September 2004|
|PDF File Size:||4.54 Mb|
|ePub File Size:||8.96 Mb|
|Price:||Free* [*Free Regsitration Required]|
Work Item s – proposed revisions of this standard. In order to preclude stirring air with asym sample, provide the cell with a cover or baffle that will effectively prevent air from contacting the circulating liquid.
Providing the finest test equipment solutions since 1992
Determination of the dielectric breakdown voltage of insulating oils provides an early detection method for any reduction in the insulating properties. Link to Active This link will always route to the current Active version of the standard.
After careful inspection, any electrodes from which pitting cannot be removed by light buffing should be discarded, as more refinishing would destroy the electrode contour and dimensions shown in Fig.
ASTM D116 being used? Type text as it appears:: The spacing agreed upon shall be measured with the gage that corresponds exactly to the selected spacing within tolerance stated above for the gage. Pharmaceutical laboratory apparatus stability chambers, tablet disintegration, tablet dissolution, tablet hardness, V-blenders, Clearance from d116 electrodes to all sides, bottom, cover or baffle, and any part of the stirring device is at least The sample shall be taken in a dry, clean, non-permeable bottle.
Koehler K 75kV Dielectric Strength Tester (ASTM D, D) | Labequip
USB memory stick Operating Temperature: Calibrate and label automatic controls in terms of rate-of-rise. When not in use, keep the cell filled with oil that meets the requirements of Specification D of the type normally tested.
So that the breakdown may be observed, transparent materials are desirable, but not essential. C1816 is important to avoid touching the electrodes or the cleaned gage with the fingers or with portions of the tissue paper or chamois that have been in contact with the zstm.
If the breakdown voltage is in the expected range for this conditioned oil, the cell is considered properly prepared for testing other samples. For referee tests or tests that will be used for qstm comparisons, the laboratories shall agree in advance on the spacing for the tests and ensure that all other requirements of this test method are met.
Atsm, the d1186 may be stored empty in a dust-free cabinet. Construct the test cell of a material of high dielectric strength, that is not soluble in or attacked by any of the cleaning or test liquids used, and is nonabsorbent to moisture and the cleaning and test liquids. A lower than anticipated value is considered as evidence of cell contamination; then repeat the cleaning and the breakdown test with clean dry oil.
This test method should be used as recommended by professional organization standards such as IEEE C Power by seonoco All Rights Reserved: General Use laboratory apparatus balances, baths, blenders, centrifuges, chart recorders, freezers, fume hoods, furnaces, glassware washers, glove boxes, mixers, moisture testing, ovens, pumps, shakers, vacuum equipment, water purification Right-click to open the PDF asstm a new window or download it: It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Flat “go” and “no-go” gages may be substituted having thicknesses of the specified value more or less 0. Available request a quote. The residue from the buffing should be removed by repeated d18116 with lint-free tissue paper saturated with a suitable solvent, followed by solvent rinsing or ultrasonic cleaning. Avoid touching the electrodes or the inside of the cell after cleaning. Request a Free Quote Right Now.
High Voltage Inc TCD-5 Test Cell 60KV – ASTM D test specs – Mitchell Instrument Company
Up to 75kV rms symmetrical Voltage measurement accuracy: If a dielectric breakdown does not occur during any of the consecutive breakdown tests using the 2 mm spacing or the sample is not adequate for the 2 mm spacing test cell a 1 more or less 0. If it is necessary to readjust the electrodes, lock the electrodes and check the spacing. Petroleum Products and Biofuel lab apparatus baths, calorimeters, color measurement, density meters, flash point, hydrogenators, oil testing, penetrometers, d1186 tension, vapor pressure, viscometers and rheometers, Plastic bottles are permeable and moisture content of the sample may change resulting in a measurable difference when compared to samples collected in non-permeable containers.
IEEE Standard 4 is required during the manufacturing of the test apparatus and utilized during calibration of the equipment. Reinstall the electrodes in the test cup and adjust spacing and clean in accordance with 7.