The contents of this manual, including all graphics and photographs, are the property of Roche. Diagnostics. No part of this document may be reproduced or. Then, as necessary, refer to the User Manual: The CoaguChek XS System User Manual is a comprehensive guide to the meter and test strips. It is designed to. Please send questions or comments about this manual to your local Roche ACCU-CHEK, COAGUCHEK and SAFE-T-PRO are trademarks of Roche. All other.
|Published (Last):||6 September 2016|
|PDF File Size:||7.81 Mb|
|ePub File Size:||20.51 Mb|
|Price:||Free* [*Free Regsitration Required]|
Got it, continue to print. You coauchek, however, erase individual test results. For more information, see re-insert it. Page 99 The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- x of the CoaguChek XS monitor for both capillary and venous whole blood samples is very good. Repeat the test using a new test Error: Clean the meter whenever Cleaning the Test Strip Guide it looks dirty or, if you prefer a regular schedule, clean the meter each time you open a new box of test strips.
Using Venous Blood 2. For the capillary testing, the investigator performed a duplicate fingerstick; the first drop was applied to one CoaguChek XS monitor and the second drop was applied to a second CoaguChek XS moni- tor. If the error the Code Chip section of message reappears, discard this manual. About this Manual Memory Get Ready Take a test strip out of the then press M container. There is a potential risk of infection.
Tables 1 and 2; a visual representation is provided in Coaguchej 1 and 2. Gather Items Gather the following items: Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test. Allow to Dry With the cover off, allow the The CoaguChek XS Meter includes advanced features that you can test strip guide to dry for set up as necessary. The correlation was 0.
Page of Manjal. Match the Code Confirm that the number Performing a Test displayed matches the number on the test strip container, 1.
Roche CoaguChek XS User Manual
Page 91 For the capillary testing, the investigator performed a duplicate fingerstick; the first drop was applied to one CoaguChek XS monitor and the second drop was applied to a second CoaguChek XS moni- tor.
The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole blood samples is very good.
If the memory is result in memory. Page 24 For more information, see re-insert it.
Erasing Stored Test Results 2. Code Chip Code Chip 2. If the numbers are different, Close the container tightly. The correlation coefficient is 0.
Fill the Capillary Tube When the flashing test strip 4. Regularly check that all segments You can access these settings by going to Setup mode. View Most Recent Result The most recent test result appears. Indicates the results are displayed in Seconds. Cleaning the Meter It is important to keep the mabual clean. To exit Erase xa without To erase all stored results: Blood 1 hanging drop of blood may be collected directly using a butterfly or a needle and syringe.
Table Of Contents User Manual Operating Conditions Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test. By pressing ‘print’ button you will print only current page.
Preparing For A Test 1. This graph shows manial mean bias for INR results; the overall mean bias was 0. Enter text from picture: Keep the tube level, and allow it to fill by capillary action.
ROCHE COAGUCHEK XS USER MANUAL Pdf Download.
Wurster M, Doran T. Then touch the capillary tube to the blood drop. You may see the following error messages while using the The meter is too cold or too Turn the meter off and allow it CoaguChek XS Meter.
Page 92 Tables 1 and 2; a visual representation is provided in Graphs 1 and 2. Advanced Features Advanced Features 3.
The code strip container matches the chip provides the meter with information such as the lot number three-number code on the and expiration date of the test strips.