This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3.
|Published (Last):||8 October 2005|
|PDF File Size:||7.1 Mb|
|ePub File Size:||14.71 Mb|
|Price:||Free* [*Free Regsitration Required]|
Front-panel keys that are white perform an immediate action rather than bringing up a menu.
E4440A PSA Spectrum Analyzer, 3 Hz to 26.5 GHz
Set the frequency span to 50 MHz: The following six models are available: Agilent handheld spectrum analyzer user’s guide pages. Making a Basic Measurement This chapter provides information on basic analyzer operation.
Directories are displayed as DIR and drives have nothing listed under Type:. Copying a File 1. Installation and Setup Turning on the Mqnual for the First Time If using LAN, set the IP address of the analyzer to an appropriate number for your network one that the network recognizes, but that is not yet in use: Also, listen for internal fan noise to manuql if the analyzer cooling fans are running.
Copying A File Copying a File 1.
Then press the desired terminator either a unit softkey, or the Terminate a value that requires a unit-of-measurement. In Case of Difficulty Before calling Agilent Technologies, however or returning an analyzer for serviceperform the quick checks listed in the problem. Available only at time of purchase Chapter 5 Manul Provides the factory calibration test data and the standard commercial calibration certificate at initial shipment.
If a problem persists, you may choose to: You must now press the NOTE You need to use a filename that does not already exist in the current directory. Viewing Catalogs and Saving Files The analyzer stores and retrieves data similarly to the way that a personal computer PC does: Don’t have an account? Viewing a Signal 1. When you have completed the printer setup for the analyzer, press the front panel key.
Agilent Technologies E4440A Getting Started Manual
Protecting Against Electrostatic Discharge Electrostatic discharge ESD can damage or destroy electronic components the possibility of unseen w4440a caused by ESD is present whenever components nanual transported, stored, or used. Place a marker labeled 1 on the 10 MHz peak, as shown in Press Peak Search Note that the frequency and amplitude of the marker appear both in the active function block, and in the upper-right corner of the screen.
Front and Rear Panel Features This chapter gives you an overview of the front and rear manua of your analyzer. The front-panel keys are dark gray, light gray, green, or white in color. Knob Increments or decrements the current value.
Page 18 When you have completed the printer setup for the analyzer, press the front panel key. Creating A Directory b.
The e44440a is located on the CPU board. Viewing Traces 1, 2, and 3 Step 2. Index personality options not in instrument power amplifiers preamplifiers preset factory types user, creating printer setup probes AC and DC product information rear panel symbols rear panel features reference level, setting reference, turning on 10 MHz rename files RF limiters This setting enables the alignment routine to run automatically either every 24 hours, or when the internal analyzer temperature changes 3 C.
AGILENT TECHNOLOGIES EA GETTING STARTED MANUAL Pdf Download.
Press If the previously performed alignments did not resolve the problem, press Alignments Restore Align Defaults Is e4440aa analyzer displaying an error message?
Viewing Traces 1 and 2 7. To access the Save menu, press, 2. Select the type of file you want to load setup, state, trace, limits, or corrections. Select the type of file you want to rename setup, state, trace, limits, screens, or corrections.
NOTE You should notice that the trace you saved, in this example trace 3, is loaded to trace 2. To access the menu, press Rename 2. Factory Preset Restores the analyzer to a user-defined state.
Page 30 Front and Rear Panel Features Key Overview In other key menus, one key label will always be highlighted to show which key has been selected but the e440a is immediately exited when a selection is made.
The analyzer was shipped from the factory with the Alignments mode set to Auto. Improving Frequency Accuracy The only green key is the key, which performs an analyzer reset A summary of all front Preset By pressing ‘print’ button you will print only current page.
Used Agilent EA
Page 34 Making a Basic Measurement This chapter is divided into the following sections: Make sure there is no disk in the front-panel disk drive. CAUTION The analyzer is shipped with a a transportation disk inserted in the disk drive to prevent damage to the disk drive during transportation. Making a Basic Measurement This chapter is divided into the following sections: Front and Rear Panel Features Key Overview In other key menus, one key label will always be highlighted to show which key has been selected but the menu is immediately exited when a selection is made.
Agilent Technologies shall not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance, or use of this material. Arrow Keys Numeric Keypad Enters a specific value. Note that the file extension is always set by the system and for this file type the extension is TRC.
To see that trace 2 is in view mode, blank traces 1 and 3 by pressing. Save the file 1. In Case of Difficulty This chapter includes information on how to check for a problem with your Agilent Technologies spectrum analyzer, and how to return it for service.